FREE FOR MEMBERS
Individual membership and corporate sponsorship in the Vibration Institute includes
Become a Vibration Institute Member today for only $95 and get access to the entire document library at no charge.
BUY IT FOR $99
Only interested in this article? You can make a one time purchase to download this content.
Recommended Practices For A Bump Test
by Jianke Wang
This paper is an excellent summary of do’s and don’ts for those conducting bump or natural frequency tests. Topics covered include proper hammer tip selection, required instrumentation & plotting preferences, but many other general tips learned by experience in the field are shared. The concept of coherence is well defined. The proper selection of the transfer functions, inertance (g/lbf), mobility (ips/lbf), or dynamic compliance (mills/lbf), is described. The paper concludes with a detailed discussion of results from actual natural frequency test data from a real machine (shale shaker).
“Although the bump test is one of the most fundamental methods used to determine the natural frequencies of a structure, not much literature is available regarding the detailed procedures and techniques required for an accurate reading of natural frequencies. A series of recommended practices for conducting a successful bump test are summarized in this article. An example illustrates their application for the accurate assessment of natural frequencies.
If an exciting frequency matches a structural natural frequency, resonance occurs. At resonance, vibration amplitude can be controlled only by damping; however, artificially-introduced damping is not preferred because it consumes energy. In general, resonance should be avoided in any system.
In addition, structural natural frequencies should be separated from potential exciting frequencies. A 15% separation margin is commonly recommended  although a 20% margin is advisable in some specific applications; e.g., the special purpose gear unit in API Standard 613 . An accurate reading of structural natural frequencies of a system with a bump test is a prerequisite to any code-compliant system design or design modification.
Instrumentation for a Bump Test
The following instrumentation is required to carry out a bump test.
Careful selection of a suitable hammer and tip is important for a successful bump test. In general, a softer (often also larger) hammer tip provides higher amplitudes of exciting frequency components of the impact force, a narrower force spectrum, and a lower exciting frequency. Figure 1 contains a series of sample frequency response curves for an impact hammer with tips of different size and hardness. For a specific hammer, a detailed frequency response curve supplied by the manufacturer is needed to choose a suitable hammer and tip for the proposed impact test.
"Recommended Practices for a Bump Test:
These practices for implementing a successful bump test result from extensive research and discussions with practicing vibration specialists.
Vibration Analysis Certifications
The Vibration Institute Training Courses provide unique opportunities to study vibration principles in a way that goes beyond the textbook to provide real-world applications.